|Place of Origin:||China|
|Minimum Order Quantity:||1pc|
|Packaging Details:||in wooden box|
|Delivery Time:||7 working days after received your payment|
|Payment Terms:||T/T or western Union|
|Supply Ability:||100pcs per month|
|Product Name:||Gasoline Stainless Steel / Compact Type Electromagnetic Flow Meter||Model:||MTF-FAP|
|Output Signal:||4-20mA,pulse,RS485||Liner:||Hard Rubber|
|Electrode:||316L,Hastelloy,Ti,Ta,PT||Power:||220VAC, 110VAC, 24VDC|
Gasoline Stainless Steel Electromagnetic Flow Meter with Rubber Linging
The reason for choosing us
1). It has the function of infrared remote control and button operation so as to make
the operation more convenient.
2). It provides the function of zero correction of sensor and automatic zero adjustment.
3). It has multi-language interface to satisfy all the users’ requirements.
4). It has the function of self-adjustment and self-checking.
5). A variety of output communication interfaces are optional.HART communication (optional);
RS485 (Modbus is optional) communication interface; Profibus (optional)
16-figure current loop4~20mA output 0-10kHz frequency output Pulse equivalent output.
Instrument detection and maintenance
Electromagnetic flow converter has self-diagnosis function. In addition to the
power and hardware circuit fault, The system also give alarm to other faults
in general usage. These information are given accordingly prompts in the
lower right of the screen.
1, Maintenance and repair
a) Sensor installation site should be consistent with the section four
requirements, Please keep the housing clean and tidy;
b)Transmitter should be placed in a clean, ventilated, dry place;
c)Need to inspect every two years, for high precision users, instrument
need to be submitted to censorship.
2, Transportation, storage
a)The instrument must be packed before transportation, be carefully
handled when loading and unloading.
b)Storage site must be dry, ventilated, avoid corrosive gas erosion, the
ambient temperature is not too low or too high;
c)Storage time better not exceed three years.
3, Unpacking and checking
a) The packing can’t be opened with heavy hammer, and should be
careful not to damage the instrument
b) Check the content carefully according to the packing list.
Selectrion of Ground Ring
If the pipe connecting to the meter is insulated,the ground ring is needed, generally
metallic or non-metallic grounding rings or grounding electrode (non-metallic) material is
available to adapt to the corrosion of the measured material.
If the measured medium is abrasive, ground ring with neck should be used to protect
lining material of two sides and to prolong its life.
In order to make the reliable operation of the instrument, and to improve measurement
accuracy, free from outside interference of Send potential,the sensor should have separate
ground wire, and the ground ring of both sides of the sensors should be fitted when the
ground resistance, in the pipe connecting to the sensor or non-metallic pipe, is less than 10Ω.
MTF series magnetic flow meter selection table
|Type of instrument||MTF|
|Installation form||-S||Flange-type sensor|
|Electrode form||-F||Standard and fixed type|
|Electrode material||A||Stainless steel 316L|
|O||Please mark out others|
|Measuring pipe material||B||304 Stainless steel (Standard)|
|C||316 Stainless steel|
|Flange Material||A||304 stainless steel|
|B||316 Stainless steel|
|C||20# Carbon steel|
|Cover Material||D||Cover 304 stainless steel|
|E||Cover 316 Stainless steel|
|F||Cover Carbon steel paint (Epoxy metallic paint)|
|Installation material flange||0||Without matching flange|
|1||With matching flange|
|Ground ring||0||Without ground ring|
|1||With ground ring|
|Form of converter||-I||Integrated type|
|Output signal and communication||P||Pulse|
|Power supply||0||220V AC|
|Explosive proof level||0||Non-explosive proof|