|Place of Origin:||China|
|Minimum Order Quantity:||1pc|
|Packaging Details:||in wooden box|
|Delivery Time:||7 working days after received your payment|
|Payment Terms:||T/T or western Union|
|Supply Ability:||100pcs per month|
|Product Name:||Stainless Steel Electromagnetic Flow Meter For Disposal Industry||Application:||Environmental Protection、Chemical Industry Etc.|
|Pressure:||ANSI Class 150||Flange:||SS304|
|Protection Level:||IP68||Output Signal:||4-20mA, Pulse|
Disposal Industry Electromagnetic Flow Meter Stainless Steel Magnetic Flowmeters
MTF electromagnetic flow meter is, a kind of instrument,used for measuring the flow
conductivity liquid, which is widely used in flow measuring of Metallurgy, Chemical
industry, Paper industry, Environmental protection, Oil industry, Textile industry, Food
industry and City administration, Water works,etc.
1). It has lightning-protection and electricity-protection circuits. The effective,
anti-interference circuits apply to all kinds of bad environment
2). It has the function of self-adjustment and self-checking.
|3). Easy for installation, 5D for front straight pipe, 2D for back straight pipe(D means inner diameter of chosen instrument)|
|4).High accuracy, it is up to ±0.2%,it is capable for testing|
|5). It provides the function of zero correction of sensor and automatic zero adjustment.|
|6). It has multi-language interface to satisfy all the users’ requirements.|
Option of the Instrument Diameter
The diameter of the meter is not necessarily the same size as the diameter of pipe,
it should be determined by the flow rate.
Generally,the flow rate between 0.5-5m/s is called economic flow rate, and the diameter
of sensor is the same size as the diameter of pipe. Unless the medium gets a less abrasion,
and use the abrasion-resistant medium, otherwise we recommend it would not exceed 7m/s
for long-term working.
For the flow contains material of adhering, precipitating and scaling,we recommend flow
of 2m/s, preferably up to3-4m/s or more to achieve self-cleaning and prevent adhering and scaling;For those strong abrasive fluid like slurry,the common flow should be less than 3m/s
to reduce the abrasion to the lining and electrode; for the low conductive medium within
5-10μ/S,the flow of 0.5-1m/s would be better.Du to the increase of the flow rate will increase
the flow noise,the output shake will appear.
Selection of Electrode materials